KODAK POC 360 CR IMAGE QUALITY
November 14, 2019 09:04
HELLO , the CR image looks dragged and cut off. I wonder if it is calibration issue or mechanical problem? Please advice for possible solutions. Every help will be much appreciated. I can send an example xray pic for you to see what I mean.
to rate and post a comment
re: KODAK POC 360 CR IMAGE QUALITY
November 15, 2019 09:31
To be expected from an Orex digitizer, I mean ScareStream.
Remove cover and inspect tension springs on feed rollers, they have a tendency to break causing the plate return spring to slide the plate backwards.
Basically the design is absolute junk and hasn't changed from the days of Orex.
If springs look good you could try to do an image center calibration.
Special-Pricing Available on Medical Displays, Patient Monitors, Recorders, Printers, Media, Ultrasound Machines, and Cameras.
This includes Top Brands such as SONY, BARCO, NDS, NEC, LG, EDAN, EIZO, ELO, FSN, PANASONIC, MITSUBISHI, OLYMPUS, & WIDE.