CPI X RAY GENERATOR Installation of TUBE
March 16, 2020 10:49
i have Two CPI Genarators INDICO 100 RAD 50 KW with RAD Console .
and two tube. one is SHIMADZU tube U10CN. and second one Toshiba E7252X .
what are the sitting for Shimadzu and Toshiba tube with CPI Generator.
please help me if possible. any one who has practical experience on CPI GENERATOR it.
to rate and post a comment
re: CPI X RAY GENERATOR Installation of TUBE
March 17, 2020 10:39
Do you have the service manual for the Indico 100 and know how to enter the service mode for calibrations/programming?
re: re: CPI X RAY GENERATOR Installation of TUBE
March 19, 2020 10:44
You will need the genware service software
re: re: re: CPI X RAY GENERATOR Installation of TUBE
March 23, 2020 10:29
Thanks for support . i have genware service software.
yes i have all the service manual. but the tube model is not found . that why i need help that what shoud i do . my tube shadzmo U10CN.
Hi, you could choose anyone with the same focal spots, power and speed. And, of couse, calibrate them. If you need manuals let me know.
March 17, 2020 10:43
i am tried before to test shimadzu x ray tube with the CPI generator, but it cause defect the rotor board,
so if you need to install tube , install the Toshiba E7239 its normal speed,
but the E7252 its high speed , not recommended,
thanks sir . if i low the Roter speed does it work will sir .
15,000 service professionals from over 50 different countries have trained in over 35 courses in radiology, mammography, CT, ultrasound, networking, PACS, & DICOM. Classroom & lab facilities to accommodate up to 60 students.