Here is a Leica DM IL Inverted Phase Contrast Microscope Pred DMIL LED Unit2
Setup for Brightfield with Phase Contrast
This Leica DM IL Inverted Phase Contrast Microscope is in good working order with all aspects tested and fully working.
Microscope optics have been cleaned professionally by our technician under a class 10,000 clean room hood.
Please see the exact video and photos below.
Testing we performed:
The Leica DM IL Inverted Phase Contrast Microscope passed all tests and was found to be fully functional:
Tested initial power to the microscope: PASSED
Tested halogen lamp and adjustment: PASSED
Tested camera and binocular ports: PASSED
Tested Z-axis coarse and fine focus: PASSED
Tested mechanical XY stage: PASSED
Tested mechanical Z-axis: PASSED
Tested brightfield light path: PASSED
Tested objective for clarity: PASSED
Tested MU300 camera live feed and clarity: PASSED
During testing and cleaning this Leica DM IL Inverted Phase Contrast Microscope was found to have no issues.
A clear image was produced through the eyepieces as well as through the camera port.
All of the mechanical components moved freely and the test sample was able to be focused without excessive effort.
System consists of:
Leica DM IL Inverted Microscope
Trinocular Tube Head
Phase Contrast Slide Ph1
6V 35W Halogen Lamp
Mechanical XY Stage
Mechanical 4-position Turret
10x/0.25 N PLAN
20x/0.30 C PLAN L Ph1
Photo/ Port Splits:
100% Eye or 100% Camera
AmScope MU300 CMOS 3.1MP Digital Color Camera
Dell Latitude i5 running Windows 10 with AmScope software
Communication and power cables.
Configuration and Setup:
Illumination Type: Brightfield w/ Phase Contrast
Objectives: 10x/0.25 N PLAN, 20x/0.30 C PLAN L Ph1
Light Sources: 6V 35W Halogen Lamp
Stage Type: Mechanical XY
This Leica DM IL Inverted Phase Contrast Microscope is currently set up for Brightfield and Phase Contrast studies.
The MU300 AmScope CMOS digital color camera is capable of taking detailed digital images.
The Leica DM IL Inverted Phase Contrast Microscopeincludes a Dell Latitude i5 laptop to control the AmScope software for live feed images.
This video begins with a full front view of the Leica DM IL Inverted Phase Contrast Microscope and all components powered up and running.
The camera scans from one side to the other for a closer look. Next, we see the AmScope live feed, focus, stage XY and light intensity.
TO VIEW ADDITIONAL PHOTOS, VIDEOS AND VERY DETAILED INFORMATION SUCH AS TEST STEPS PERFORMED, CONFIGURATION, ETC., PLEASE VISIT OUR WEB SITE. TO GET THERE PLEASE CLICK ON "WEBSTORE" TO YOUR RIGHT AND YOU WILL SEE LINK TO OUR SITE IN THE MIDDLE OF THE SCREEN. THEN USE LAST 5 DIGITS OF MPN NUMBER IN THE SEARCH FIELD TO LOCATE YOUR ITEM.
DO NOT HESITATE TO CALL US OR EMAIL US WITH ANY QUESTIONS YOU MIGHT HAVE.
ALL OF OUR ITEMS ARE AVAILABLE FOR INSPECTION AT OUR FACILITY.
Items are sold as-is with no returns or refunds available unless explicitly stated.