Here is a GE AKTA FPLC Detector UV-900 Monitor UV900 10mm flow cell
This GE AKTA FPLC Detector UV-900 Monitor is in good working condition.
Please see the exact photos and video below.
EACH UNIT CONTAINS MANY WARRANTY STICKERS TO PREVENT PARTS FROM BEING SWAPPED, INCLUDING ON ALL INTERNAL BOARDS. WE WILL NOT ACCEPT ANY UNITS BACK IF ANY OF THE STICKERS ARE CUT OR DAMAGED. WE APOLOGIZE ABOUT THIS MESSAGE BUT WE HAD AN OCCURRENCE WITH CUSTOMER ATTEMPTING TO REMOVE DEFECTIVE UNIT BACK AFTER SWAPPING PARTS.
Testing we performed:
The GE AKTA FPLC Detector UV-900 Monitor passed all tests and was found to be fully functional:
Prom test: PASSED
RAM test: PASSED
EEPROM test: PASSED
Display test: PASSED
Key test: PASSED
Dial test: PASSED
Uninet 1&2 test: PASSED
Buzzer test: PASSED
Block filter test: PASSED
Digital inports test: PASSED
Lamp intensity test: PASSED
Light intensity test: PASSED
Noise test (10 loops): PASSED
Stepper motor test: PASSED
Dark current test: PASSED
Motor stress test: PASSED
System consists of:
GE AKTA FPLC Detector UV-900 Monitor
10mm flow cell
Flow cell holder plate
Standard power cord.
Configuration and Setup:
Detector type: UV/Vis monitor
Max. flow rate: 100 ml/min, Max. pressure 2 MPa (20 bar, 290 psi), Backpressure Max. 0.5 bar at 2 ml/min
Light source: Xenon flash lamp, Lamp lifetime > 4000 hours, Control Stand alone or from a computer running, UNICORN 3.21 or higher through, UniNet-1 connection
Wavelength range: 190â700 nm in steps of 1 nm, 3 wavelengths simultaneously, Bandwidth 4 nm, Wavelength accuracy +/-2 nm, Wavelength reproducibility +/-0.01 nm, Wavelength switch time < 500 ms (one cycle from 214 nm to
254 nm and back to 214 nm), Linearity <2% deviation up to 2 AU at 260 nm with, Uracil at pH 2
Noise: (specified for 10 mm flow cell), 1, 2 Single wavelength short term (0.5â1 min) long term (1â10 min), Dual wavelengths: short term (0.5â1 min), long term (1â10 min) < 6x10-5 AU at 230 nm, < 6x10-5 AU at 230 nm, < 2x10-4 AU at 230 and 254 nm
< 2x10-4 AU at 230 and 254 nm Drift 2) < 2x10-4 AU/h at 254 nm
Monitored changes in Absorbance with Degassed DI water and DI water with NaI added and registered changes on screen.
We zoom out to get a look at the front of the GE AKTA FPLC Detector UV-900 Monitor, from here we pan around doing a full external inspection, paying special attention to the serial tags and connections.
Once we return to the front, we cycle through several menus to verify functionality. We return to the display and set the unit to run.
Distilled water is shown in the readings and water with an NaI is introduced.
Once complete, we zoom out for a final look.
TO VIEW ADDITIONAL PHOTOS, VIDEOS AND VERY DETAILED INFORMATION SUCH AS TEST STEPS PERFORMED, CONFIGURATION, ETC., PLEASE VISIT OUR WEB SITE. TO GET THERE PLEASE CLICK ON "WEBSTORE" TO YOUR RIGHT AND YOU WILL SEE LINK TO OUR SITE IN THE MIDDLE OF THE SCREEN. THEN USE LAST 5 DIGITS OF MPN NUMBER IN THE SEARCH FIELD TO LOCATE YOUR ITEM.
DO NOT HESITATE TO CALL US OR EMAIL US WITH ANY QUESTIONS YOU MIGHT HAVE.
ALL OF OUR ITEMS ARE AVAILABLE FOR INSPECTION AT OUR FACILITY.
Items are sold as-is with no returns or refunds available unless explicitly stated.